You’ll get there quickly. Message 6 of The driver only uses the Fetch and Read commands so I can’t confirm that the Measure command is correct. I have the HPA Init. I followed your conversation with Joe For the measurements, go back a couple of steps and select “Measurement”, there are a couple of selections to choose from. Remote operation of the instrument usually:
|Date Added:||6 January 2008|
|File Size:||68.58 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Then I can get the curve U t and I t. For the measurements, go back a couple of steps and select “Measurement”, there are a couple of selections to choose from.
So try the following: Message 5 of Unfortunately, I did not see any examples or “getting started” vi’s for this one. Hello Joe and Anja, Thank for your help.
I don’t understand the attachment. You’re not using the driver for the instrument.
If you got this far, double click “Configure” and place the vi onto your block diagram. First I might suggest to write in german as that might be easier for you?
Message 9 of In this way you can communicate with every GPIB-instrument easily without the absolut need of an instrument driver.
And are you sure of the syntax for the measure command? See attached picture for additional functions: The same as the structure in Attachments. Thank you very much for your attention.
I also sometimes ignore existing drivers and create my own. I have the HPA Init. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. I admit you might be faster with an existing driver and of course you are more sure you didn’t forget something important but I sometimes had instruments without existing drivers.
You’ll get there 344001a.
A GPIB VISA TIME OUT ERRORS | Keysight Community
Of course you can add some kind of control of the communication if you want. Message 3 of Message 2 of They are embedded inside the instrument driver selections. Most Active Software Boards: You can now ask the instrument for the currently measured data i. First thing is to obtain the LV drivers for the A.
Let’s see if I can get you on your way with the following: The front panel wi ll allow you to select the function you want to do, including DC current.
Message 8 of Thank you Very much. Just refer to your instrument manuals section for programming. I never had problems with this till now.
How can i do with these?